Miniscrews in orthodontic treatment: Review and analysis of published clinical trials
A systematic review of effects related to patient, screw, surgery, and loading on the stability of miniscrews was conducted.
Reports of clinical trials published before September 2007 with at least 30 miniscrews were reviewed. Parameters examined were patient sex and age, location and method of screw placement, screw length and diameter, time, and amount of loading.
Fourteen clinical trials included 452 patients and 1519 screws. The mean overall success rate was 83.8% + or - 7.4%. Patient sex showed no significant differences. In terms of age, 1 of 5 studies with patients over 30 years of age showed a significant difference (P <0.05). Screw diameters of 1 to 1.1 mm yielded significantly lower success rates than those of 1.5 to 2.3 mm. One study reported significantly lower success rates for 6-mm vs 8-mm long miniscrews (72% vs 90%). Screw placement with or without a surgical flap showed contradictory results between studies. Three studies showed significantly higher success rates for maxillary than for mandibular screws. Loading and healing period were not significant in the miniscrews' success rates.
All 14 articles described success rates sufficient for orthodontic treatment. Placement protocols varied markedly. Screws under 8 mm in length and 1.2 mm in diameter should be avoided. Immediate or early loading up to 200 cN was adequate and showed no significant influence on screw stability.
Available from: Yuchin Wu
- "With the exception of five TSADs (two failed between 0 and 4 weeks and three TSADs became loose between 4 and 8 weeks), the remaining twenty were stable and well integrated with the surrounding alveolar bone during the 8 week experimental period. Our success rate of 92% at 4 weeks and 80% at 8 weeks was comparable to a success rate of 83.6% reported in recent meta-analyses [18, 19]. Various factors affect TSAD stability during insertion and following loading and can be broadly divided into factors affecting primary and secondary stability. "
[Show abstract] [Hide abstract]
ABSTRACT: The aim of this animal study was to develop a model of orthodontic tooth movement using a microimplant as a TSAD in rodents. A finite element model of the TSAD in alveolar bone was built using μCT images of rat maxilla to determine the von Mises stresses and displacement in the alveolar bone surrounding the TSAD. For in vivo validation of the FE model, Sprague-Dawley rats (n = 25) were used and a Stryker 1.2 × 3 mm microimplant was inserted in the right maxilla and used to protract the right first permanent molar using a NiTi closed coil spring. Tooth movement measurements were taken at baseline, 4 and 8 weeks. At 8 weeks, animals were euthanized and tissues were analyzed by histology and EPMA. FE modeling showed maximum von Mises stress of 45 Mpa near the apex of TSAD but the average von Mises stress was under 25 Mpa. Appreciable tooth movement of 0.62 ± 0.04 mm at 4 weeks and 1.99 ± 0.14 mm at 8 weeks was obtained. Histological and EPMA results demonstrated no active bone remodeling around the TSAD at 8 weeks depicting good secondary stability. This study provided evidence that protracted tooth movement is achieved in small animals using TSADs.
Available from: maneyonline.com
- "A systematic review by Reynders et al. (2009) identified 19 studies of sufficient quality and the success rate was mainly in excess of 80%. Another review by Crisman et al. (2010) included 14 studies and reported an overall success rate of 84% in 1,519 screws. "
Available from: Manuel Nienkemper
- "Their versatility has made new types of mechanics and treatment options possible.12,13 Regarding orthodontic mini-implants, current meta-analyses have reported a success rate of 83.6%.14,15 "
[Show abstract] [Hide abstract]
ABSTRACT: A growing number of studies have reported that mini-implants do not remain in exactly the same position during treatment, although they remain stable. The aim of this review was to collect data regarding primary displacement immediately straight after loading and secondary displacement over time. A systematic review was performed to investigate primary and secondary displacement. The amount and type of displacement were recorded. A total of 27 studies were included. Sixteen in vitro studies or studies using finite element analysis addressed primary displacement, and nine clinical studies and two animal studies addressed secondary displacement. Significant primary displacement was detected (6.4-24.4 µm) for relevant orthodontic forces (0.5-2.5 N). The mean secondary displacement ranged from 0 to 2.7 mm for entire mini-implants. The maximum values for each clinical study ranged from 1.0 to 4.1 mm for the head, 1.0 to 1.5 for the body and 1.0 to 1.92 mm for the tail part. The most frequent type of movement was controlled tipping or bodily movement. Primary displacement did not reach a clinically significant level. However, clinicians can expect relevant secondary displacement in the direction of force. Consequently, decentralized insertion within the inter-radicular space, away from force direction, might be favourable. More evidence is needed to provide quantitative recommendations.International Journal of Oral Science (2013) 5, doi:10.1038/ijos.2013.92; published online 13 December 2013.
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.