[Show abstract][Hide abstract] ABSTRACT: Using a scanning microwave microscope (SMM), we have investigated the phase separation in a 30% La <sub>5/8</sub> Sr <sub>3/8</sub> Mn O <sub>3</sub> ( LSMO )+70% Lu Mn O <sub>3</sub> ( LMO ) polycrystalline pressed powder sample, in which the LSMO phase is a perovskite ferromagnetic metal while the LMO phase is a hexagonal ferroelectric insulator. When the electrical properties of the sample were imaged using our SMM, the sample showed a significant contrast between the metallic LSMO and the insulating LMO grains, indicating a clear phase separation between the two phases. The metallic phase identified by the SMM clearly showed a ferromagnetic signal when investigated by a magnetic force microscope (MFM), providing solid evidence that the metallic phase is indeed the ferromagnetic LSMO. In addition, we have noticed a slight difference between the images generated by SMM and MFM, and we believe that this is due to the different depth scales probed by the two microscopy techniques.
[Show abstract][Hide abstract] ABSTRACT: A scanning microwave microscope (SMM) is a microscope which uses
microwave as a media. And it can measure the dielectric constant and the
conductivity of dielectric samples simultaneously. By modifying the
electronics, an SMM can be made to acquire the topological data of a
sample surface while measuring the local conductivity. The scanned image
of the surface of a coin verifies its operation. With measuring a
reference sample, it will be shown that the vertical resolution of the
microscope is certified as less than 0.05 μm by showing our image
taken on our reference sample.
[Show abstract][Hide abstract] ABSTRACT: We obtained the electrical images of biological specimens using SMM. SMM investigates local electrical properties of materials in high frequency region ( 1.5 GHz) by detecting the shift of the resonant frequency and the quality factor Q of the resonator. At first, we investigated the NaCl solutions of various concentrations from the de-ionized water to the saturated solution, and observed the drastic change of the resonant frequency and Q depending on the concentration. Since the water was the dominant source of the dielectric signals in biological samples, the change of the resonant frequency and Q of the resonator could be a good reference interpreting various biological samples. We will present various high-frequency electrical images of the biological samples of a plant epidermal cell, and a bone section tissue.
[Show abstract][Hide abstract] ABSTRACT: We investigated the twin domains of LaAlO3 (001) substrate using a scanning microwave microscope (SMM). Since the SMM can image the local dielectric constant of a sample quantitatively, we can observe the difference of the dielectric constant in the twin domains. Especially the (110) domains were observed more clearly than (100) domains, we attribute this to the difference of the strain in the ferroelastic LaAlO3.
Japanese Journal of Applied Physics 11/2001; 40(11):6510-6513. DOI:10.1143/JJAP.40.6510 · 1.13 Impact Factor