[show abstract][hide abstract] ABSTRACT: To evaluate the time taken for rewarming hypothermic neonates and to correlate the time taken for rewarming with severity of hypothermia (WHO classification), weight, gestational age and associated morbidity.
100 extramural neonates transported to the Referral neonatal unit of a teaching hospital, with weight more than 1000 grams and abdominal skin temperature less than 36.5 oC at admission were included in the study. Hypothermia was classified as per WHO recommendations. Clinical features including age, weight, gestational age, clinical diagnosis and vitals were recorded at the time of admission. Rewarming was done under a servo-controlled radiant warmer, in skin mode at set temperature of 37 oC. Skin and air temperatures measured by the thermistor probe were recorded at the time of admission and then at least every 15 minutes till skin temperature reached 36.5 oC. The neonates were monitored for oxygen saturation, blood glucose and capillary filling time and stabilized promptly.
The mean abdominal skin temperature was 34.9 +/- 1.4 oC. 72% of babies were moderately or severely hypothermic as per WHO classification. The duration of rewarming was 4.9 +/- 0.8 min, 17.5 +/- 9.5 min and 42+/-7.9 min for mild, moderate and severe hypothermia respectively (p=0.021). The difference in rate of rewarming between various grades of hypothermia was also significant. The duration of rewarming a baby did not differ significantly between the different weight and gestational age groups. When the rate of rewarming was expressed as rise in oC per Kg body weight per hour, it was higher in smaller and more premature babies. The rate of rewarming was slower in asphyxiated babies.
The duration of rewarming depends on the severity of hypothermia. When rewarmed under radiant warmer using servo mode, the duration of rewarming a baby is the same irrespective of weight and gestational age. Asphyxiated babies take longer time to rewarm.
The Indian Journal of Pediatrics 06/2006; 73(5):395-9. · 0.72 Impact Factor
[show abstract][hide abstract] ABSTRACT: Serial skin (sole) and rectal temperatures were simultaneously taken from 55 healthy and 26 septicaemic newborn infants to find out prospectively whether septicaemic newborn infants have any thermoregulatory reaction to the septicaemia, and whether regular temperature measurements could help in the early diagnosis of septicaemia. The septicaemic infants were divided into three groups: the first comprised eight feverish infants, the second 11 with normal temperatures who were in relatively good clinical condition, and the third seven with normal temperatures who were in poor clinical condition. All 55 healthy babies had rectal temperatures of less than 37.8 degrees C and a mean rectal-sole temperature difference of 2.5 degrees C. The first group of septicaemic infants had rectal temperatures over 37.8 degrees C and a high mean temperature difference of 6.9 degrees C, whereas the second group had rectal temperatures less than 37.8 degrees C and a mean temperature difference of 4.7 degrees C. Infants of the third group had a low rectal temperature and a low mean temperature difference (1.1 degrees C). We conclude that septicaemic newborn infants show an adequate thermoregulatory reaction, which is reflected by a widening of the rectal-sole temperature difference of more than 3.5 degrees C, except for those who are critically ill, who lose this ability. In view of these results infants with normal temperatures but with a rectal-sole temperature difference of more than 3.5 degrees C should be suspected of septicaemia and investigated thoroughly.
Archives of Disease in Childhood 05/1990; 65(4 Spec No):380-2. · 3.05 Impact Factor
[show abstract][hide abstract] ABSTRACT: In closed incubators, radiative heat loss (R) which is assessed from the mean radiant temperature (Tr) accounts for 40-60% of the neonate's total heat loss. In the absence of a benchmark method to calculate Tr--often considered to be the same as the air incubator temperature-errors could have a considerable impact on the thermal management of neonates. We compared Tr using two conventional methods (measurement with a black-globe thermometer and a radiative "view factor" approach) and two methods based on nude thermal manikins (a simple, schematic design from Wheldon and a multisegment, anthropometric device developed in our laboratory). By taking the Tr estimations for each method, we calculated metabolic heat production values by partitional calorimetry and then compared them with the values calculated from V(O2) and V(CO2) measured in 13 preterm neonates. Comparisons between the calculated and measured metabolic heat production values showed that the two conventional methods and Wheldon's manikin underestimated R, whereas when using the anthropomorphic thermal manikin, the simulated versus clinical difference was not statistically significant. In conclusion, there is a need for a safety standard for measuring TR in a closed incubator. This standard should also make available estimating equations for all avenues of the neonate's heat exchange considering the metabolic heat production and the modifying influence of the thermal insulation provided by the diaper and by the mattress. Although thermal manikins appear to be particularly appropriate for measuring Tr, the current lack of standardized procedures limits their widespread use.
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.