Application of k-space energy spectrum analysis to susceptibility field mapping and distortion correction in gradient-echo EPI.
ABSTRACT Echo-planar imaging (EPI) is widely used in functional MRI studies. It is well known that EPI quality is usually degraded by geometric distortions, when there exist susceptibility field inhomogeneities. EPI distortions may be corrected if the field maps are available. It is possible to estimate the susceptibility field gradients from the phase reconstruction of a single-TE EPI image, after a successful phase-unwrapping procedure. However, in regions affected by pronounced field gradients, the phase-unwrapping of a single-TE image may fail, and therefore the estimated field maps may be incorrect. It has been reported that the field inhomogeneity may be calculated more reliably from T2*-weighted images corresponding to multiple TEs. However, the multi-TE MRI field mapping increases the scan time. Furthermore, the measured field maps may be invalid if the subject's position changes during dynamic scans. To overcome the limitations in conventional field mapping approaches, a novel k-space energy spectrum analysis algorithm is developed, which quantifies the spatially dependent echo-shifting effect and the susceptibility field gradients directly from the k-space data of single-TE gradient-echo EPI. Using the k-space energy spectrum analysis, susceptibility field gradients can be reliably measured without phase-unwrapping, and EPI distortions can be corrected without extra field mapping scans or pulse sequence modification. The reported technique can be used to retrospectively improve the image quality of the previously acquired EPI and functional MRI data, provided that the complex-domain k-space data are still available.
- SourceAvailable from: Wei Lin[Show abstract] [Hide abstract]
ABSTRACT: Off-resonance artifacts hinder the wider applicability of echo-planar imaging and non-Cartesian MRI methods such as radial and spiral. In this work, a general and rapid method is proposed for off-resonance artifacts correction based on data convolution in k-space. The acquired k-space is divided into multiple segments based on their acquisition times. Off-resonance-induced artifact within each segment is removed by applying a convolution kernel, which is the Fourier transform of an off-resonance correcting spatial phase modulation term. The field map is determined from the inverse Fourier transform of a basis kernel, which is calibrated from data fitting in k-space. The technique was demonstrated in phantom and in vivo studies for radial, spiral and echo-planar imaging datasets. For radial acquisitions, the proposed method allows the self-calibration of the field map from the imaging data, when an alternating view-angle ordering scheme is used. An additional advantage for off-resonance artifacts correction based on data convolution in k-space is the reusability of convolution kernels to images acquired with the same sequence but different contrasts.Magnetic Resonance in Medicine 10/2011; 67(6):1547-55. · 3.27 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: To analyze the streak artifacts in a gradient-echo echo planar imaging (GE-EPI) sequence and to propose a correction method for the Nyquist ghost artifacts that does not cause streak artifacts in the GE-EPI imaging. Several GE-EPI imaging experiments with various reference scans, using both GE-EPI and SE-EPI scan data, were performed to analyze the streak artifacts and to investigate the spin dephasing phenomena of the GE-EPI reference scan. In addition, the analysis based on the spin dephasing was undertaken in order to demonstrate that the SE-EPI reference data can be used for the correction of the GE-EPI main scan data. The experimental results confirmed that the improvement of the reference data using either signal averaging or a large flip angle cannot guarantee perfect correction of the streak artifact if the noise is not completely removed. Due to the main field inhomogeneity, the spins of the GE-EPI reference data were dephased in multiple echo signals. The proposed correction method, which uses a SE-EPI reference scan for the GE-EPI images, eliminates the N/2 ghost artifacts without producing streak artifacts. It is believed that the proposed phase error correction scheme can improve the EPI performance in high field MRIs with higher magnetic field inhomogeneities.MAGMA Magnetic Resonance Materials in Physics Biology and Medicine 11/2011; 25(3):205-13. · 1.86 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: BACKGROUND: Aberrant accumulation of transition metals in the brain may have an early and important role in the pathogenesis of several neurodegenerative disorders, including Huntington disease (HD). OBJECTIVE: To comprehensively evaluate and validate the distribution of metal deposition in the brain using advanced magnetic resonance imaging methods from the premanifest through symptomatic stages of HD. DESIGN: Observational study. SETTING: University imaging center. PARTICIPANTS: Twenty-eight HD expanded gene carriers, 34 patients with symptomatic HD, and 56 age- and sex-matched healthy control subjects were included in the study. INTERVENTIONS: Participants underwent magnetic resonance imaging for the quantification of the phase evolution of susceptibility-weighted images. MAIN OUTCOME MEASURES: To verify the identity of the metals responsible for the changes in the phase evolution of the susceptibility signal in the brain and to assess correlations with systemic levels. Inductively coupled plasma mass spectrometry was used to measure transition metal concentrations in postmortem brains. RESULTS: In the basal ganglia, progressive increases in the phase evolution were found in HD, beginning in premanifest individuals who were far from expected onset and increasing with proximity to expected onset and thereafter. Increases in the cerebral cortex were regionally selective and present only in symptomatic HD. Increases were verified by excessive deposition of brain iron, but a complex alteration in other transition metals was found. CONCLUSION: An important and early role of altered metal homeostasis is suggested in the pathogenesis of HD.Archives of neurology 03/2012; · 7.58 Impact Factor
Application of k-space energy spectrum analysis to susceptibility field mapping and distortion correction in gradient-echo EPI
N-K. Chen1, K. Oshio1,2, L. P. Panych1
1Radiology, Brigham and Women's Hospital, Harvard Medical School, Boston, MA, United States, 2Diagnostic Radiology, Keio University, Tokyo, Japan
Echo planar imaging (EPI), with its high temporal resolution, has become an important technique for various dynamic studies, such as functional
MRI and contrast agent enhanced imaging. It is well known that EPI is geometrically distorted by Bo field inhomogeneity. In order to remove the
geometric distortions, an extra field mapping scan can be performed prior to EPI based dynamic imaging (Jezzard and Balaban, MRM, 1995, 34:65).
However, the acquired field map may become invalid if the subject's position changes during dynamic EPI scans. Furthermore, the experimental time
will be increased when an extra field map scan is performed.
Here, a novel algorithm termed k-space energy spectrum analysis is presented to map the susceptibility field gradients directly from gradient-echo
EPI data. Neither extra field mapping scan nor EPI pulse sequence modification is needed. Based on the calculated susceptibility field gradient maps,
the Bo field inhomogeneity map can be derived and used to remove the geometric distortions in EPI data.
Theory and Methods
Echo-shifting effect in gradient-echo EPI: In the presence of susceptibility field gradients, gradient-echo EPI's echo signals deviate from the center
of the k-space and the degree of this echo-shifting effect depends on (1) the effective echo time, (2) echo-spacing time, (3) image field of view (FOV),
and (4) the susceptibility field gradients. If the echo-shifting effects, corresponding to different image-domain regions, can all be measured directly from
gradient-echo EPI data, the susceptibility field gradients can be quantified without an extra field mapping scan.
K-space energy spectrum analysis: We have developed a k-space energy spectrum analysis algorithm to map EPI's echo-shifting effect along both
readout and phase-encoding directions. The procedure for mapping the echo-shifting effect along the phase-encoding direction consists of the following
two steps. First, a selected number of ky lines are truncated (in post-processing) and replaced by values calculated from the un-truncated data using
Cuppen's partial Fourier technique (van Cuppen and van Est, MRI, 1987, 5:526). The modified k-space data are then Fourier transformed to an image.
The signal intensity of the reconstructed image will resemble the full-Fourier image, if the peak of the corresponding k-space signals is located in the un-
truncated area. On the other hand, if the peak of the k-space signals is located in the truncated k-space, there will be significant signal loss in the
reconstructed image. Second, by varying the number of truncated/replaced ky lines in the first step, a series of images are obtained. The pattern of pixel-
wise signal variation in those images is determined by the k-space echo shifting effect originating from the susceptibility field gradients. The measured
pixel-wise signal variation pattern is therefore termed k-space energy spectrum, in which abrupt signal changes occur when the peaks of k-space echo
signals are being truncated. For example, Figure 1a presents an EPI image and Figure 1b shows the phase-encoding k-space energy spectrum of three
selected pixels. As indicated by arrows in Figure 1b, the echo-signal peaks are located in ky line number 27, 48, and 74, respectively, as a result of
spatially varying susceptibility field gradients. The above two steps can also be applied to measure the k-space echo-shifting effect along the readout
direction, by truncating different number of kx lines in EPI data and calculating the corresponding k-space energy spectrum.
Measurement of susceptibility field gradients and EPI distortion correction: Once the spatially dependent echo-shifting effects along both readout
and phase-encoding directions are measured (Nx and Ny), the susceptibility field gradients along these two directions (Gx and Gy) can be calculated
based on Equation 1 and Equation 2, where tDW is the dwell time, TEeff is the effective echo time, and tESP is the echo-spacing time.
γ FOVx[Nx× τDW+ (TEeff+ Ny× τESP)] ......... 
For example, the phase-encoding susceptibility field gradient map calculated from the EPI data in Figure 1a is shown in Figure 1c. When the RF
excitation pulse frequency is chosen to be the averaged Larmor frequency of the imaged slice, Bo field inhomogeneity maps can be derived from the
susceptibility field gradients, and then used to remove EPI's nonlinear geometric distortions through the previously reported phase modulation method. If
the RF pulse frequency is set to off-resonance, there may remain residual global translation after nonlinear distortion correction. This remaining image
translation can easily be detected and corrected with rigid body image registration.
Human brain and phantom EPI data acquired at 1.5 T and 3 T GE systems were processed with the proposed k-space energy spectrum analysis
algorithm. The susceptibility field gradients were calculated based on Equations 1 and 2 (without the need of extra field mapping scan) and then used to
correct EPI geometric distortions through the phase modulation method (Chen and Wyrwicz, MRM, 1999, 41:1206).
Results and Discussion
Phantom EPI images acquired at 3T under three different shim settings are shown in Figure 2a. Using the proposed k-space energy spectrum
analysis and distortion correction technique, the nonlinear geometric distortions in EPI data are removed, as shown in Figure 2b. Examples of distortion
correction in human brain EPI are presented in Figure 3. As shown, there exist significant geometric distortions in human brain EPI (Figure 3a), and
these artifacts can be reliably removed using the proposed data processing algorithm (Figure 3b).
The proposed method is superior to conventional field map based EPI correction methods in three ways. First, using k-space energy spectrum
analysis algorithm, the field maps can be obtained without
performing additional field mapping scan. Second, in the
presence of subject movement during dynamic EPI scans,
the valid field maps corresponding to different scan times
can be reliably obtained with the proposed post-processing
method. Third, in the proposed algorithm, complicated in-
plane phase unwrapping procedure that is required in most
conventional field mapping methods can be avoided. It
should be noted that this EPI distortion correction
technique is also compatible with parallel imaging methods.
Even though the EPI distortions can generally be reduced
when data are acquired with parallel imaging schemes, the
residual distortions may still degrade the spatial accuracy,
especially at high field. Using the proposed post-processing
algorithm, the residual distortion in parallel EPI can be
γ FOVy(N ×τESP+ TEeff) ......... 
Proc. Intl. Soc. Mag. Reson. Med. 13 (2005)