A new test set-up for skull fracture characterisation.
ABSTRACT Skull fracture is a frequently observed type of severe head injury. Historically, a variety of impact test set-ups and techniques have been used for investigating skull fracture. The most frequently used are the free-fall technique, the guided fall or drop tower set-up and the piston-driven impactor set-up. This document proposes a new type of set-up for cadaver head impact testing which combines the strengths of the most frequently used techniques and devices. The set-up consists of two pendulums, which allow for a 1 degree of freedom rotational motion. The first pendulum is the impactor and is used to strike the blow. The head is attached to the second pendulum using a polyester resin. Local skull deformation and impact force are measured with a sample frequency of 65 kHz. From these data, absorbed energy until skull fracture is calculated. A set-up evaluation consisting of 14 frontal skull and head impact tests shows an accurate measurement of both force and local skull deformation until fracture of the skull. Simplified mechanical models are used to analyse the different impacting techniques from literature as well as the new proposed set-up. It is concluded that the proposed test set-up is able to accurately calculate the energy absorbed by the skull until fracture with an uncertainty interval of 10%. Second, it is concluded that skull fracture caused by blunt impact occurs before any significant motion of the head. The two-pendulum set-up is the first head impact device to allow a well-controlled measurement environment without altering the skull stress distribution.
- SourceAvailable from: Angel Arias[Show abstract] [Hide abstract]
ABSTRACT: This paper deals with the mechanical behavior of polyether–ether–ketone (PEEK) under impact loading. PEEK polymers are the great interested in the field of medical implants due to their biocompatibility, weight reduction, radiology advantage and 3D printing properties. Implant applications can involve impact loading during useful life and medical installation, such as hip systems, bone anchors and cranial prostheses. In this work, the mechanical impact behavior of PEEK is compared with Ti6Al4V titanium alloy commonly used for medical applications. In order to calculate the kinetic energy absorption in the impact process, perforation tests have been conducted on plates of both materials using steel spheres of 1.3 g mass as rigid penetrators. The perforation test covered impact kinetic energies from 21 J to 131 J, the equivalent range observed in a fall, an accidental impact or a bike accident. At all impact energies, the ductile process of PEEK plates was noted and no evidence of brittle failure was observed. Numerical modeling that includes rate dependent material is presented and validated with experimental data.Composite Structures 01/2015; 124:88-99. · 3.12 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: The aim of this study was to analyze the occurrence and characteristics of orbital roof fractures, periorbital hematoma, conjunctival hemorrhage, orbital roof discoloration, and concomitant head injuries in falls from a standing height.Forensic Science Medicine and Pathology 07/2014; 10(4). · 1.96 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: Blunt force trauma forms a substantial portion of deaths worldwide. However, few studies have attempted to determine the force involved with blunt force trauma to the lateral part of the head. Nor have many studies been conducted at velocities exceeding 10m/s. The acquisition of human tissue for experimental studies is becoming increasingly difficult. As such, the current study investigates the trauma and the force involved with cranial blunt force trauma in a porcine model. Thirty whole porcine heads were subjected to single impact tests on the fronto-parietal region at velocities ranging from 10m/s to 25m/s. Half the specimens were subjected to impact by a short projectile resembling a hammer head and the other half were subjected to impact with a Hopkinson pressure bar (HPB). Both implements had the same impact diameter and were machined from the same material. The HPB is an apparatus commonly used in material testing. Its use to determine fracture force in whole cranial specimens is novel. Fractures appeared similar in both the hammer tests and HPB tests. Lacerations and fractures resembled the shape of the striker surface with the most common fracture observed being a semi-circular depressed fracture. The mean peak fracture force was 7760N (±4150N), with a mean displacement of 3.1mm (±1.1mm). Peak fracture forces concur well with previous studies although no clear trend appears to exist between level of trauma and peak impact force.Legal Medicine 08/2014; · 1.44 Impact Factor