Generic controller dedicated to telemetry-controlled microsystems
ABSTRACT This paper introduces a generic controller designed for telemetry-controlled microsystems. This controller receives a data packet through a serial link carrying a command word and the associated data, and is capable of generating a variety of control/timing signals according to the definition of the received command. The flexible microprogrammed architecture of the controller allows for defining the commands functions in an on-chip mask-programmable read-only memory.
- SourceAvailable from: wims2.org
[Show abstract] [Hide abstract]
- "The final block is the telemetry chip, which wirelessly couples the power and bidirectional data into and out of the implant. The telemetry chip designed for this microsystem ,  receives inductively coupled frequency-shift-keying-modulated clock and command data carried by an RF signal switching between 4 and 8 MHz. The received energy is used to generate regulated power for the implanted microsystem. "
ABSTRACT: Abstract-The design and performance of an integrated front end for high-channel-count neural recording microsystems is presented. This front end consists of a 3-D micromachined microelectrode array, realized using a new architecture that allows simple and rapid microassembly. A 64-site 3-D multiprobe, realized using the new architecture, interfaces with tissue volumes of less than 0.01 mm<sup>3</sup> and has a footprint of 1 mm<sup>2</sup>. For amplification, filtering, and buffering of the recorded neural signals, a custom signal-conditioning circuit provides high gain (60 dB), low noise (4.8 μV<sub>rms</sub>), and low power (50 μW) in an area of 0.098 mm<sup>2</sup>. In addition, this circuitry implements bandwidth tuning, offset compensation, and wireless gain programmability. This new approach to system integration uses a microfabricated parylene overlay cable to electrically interconnect the 3-D array and signal-conditioning circuitry. In vivo results obtained using this integrated microsystem front end in its most compact form are presented.Journal of Microelectromechanical Systems 01/2011; 19(6-19):1409 - 1421. DOI:10.1109/JMEMS.2010.2082496 · 1.75 Impact Factor