The Amplatz (R) canine duct occluder: A novel device for patent ductus arteriosus occlusion (vol 9, pg 109, 2007)
ABSTRACT The Amplatz canine duct occluder (ACDO) is a nitinol mesh device with a short waist that separates a flat distal disc from a cupped proximal disc. The device is designed to conform to the morphology of the canine patent ductus arteriosus (PDA). PDA dimensions are determined by angiography, and a guiding catheter is advanced into the main pulmonary artery via the aorta and PDA. An ACDO with a waist diameter approximately twice the angiographic minimal ductal diameter (MDD) is advanced via the catheter using an attached delivery cable until the flat distal disc deploys within the main pulmonary artery. The partially deployed ACDO, guiding catheter, and delivery cable are retracted until the distal disc engages the pulmonic ostium of the PDA. With the delivery cable stabilized, the catheter is retracted to deploy the waist across the pulmonic ostium and cupped proximal disc within the ductal ampulla. Tension on the delivery cable is released, and correct ACDO positioning and stability are confirmed by observing that the device assumes its native shape, back-and-forth maneuvering of the delivery cable, and a small contrast injection made through the guiding catheter. The delivery cable is detached and removed with the guiding catheter. To assess for any residual ductal flow, an angiogram is performed at the conclusion of the procedure, followed by Doppler echocardiography at 1 day and 3 months post-procedure. PDA occlusion in dogs with the ACDO is straightforward and extremely effective across a wide range of body weights, somatotypes, MDDs, and ductal morphologies.
- SourceAvailable from: ncbi.nlm.nih.gov[Show abstract] [Hide abstract]
ABSTRACT: A 7-month-old female, spayed border collie was referred to the Ontario Veterinary College due to a continuous murmur noted by the referring veterinarian prior to ovariohysterectomy. Auscultation confirmed a grade VI/VI continuous murmur. An echocardiogram confirmed patent ductus arteriosus (PDA). An Amplatz canine ductal occluder device was successfully placed for occlusion of blood flow though the ductus.The Canadian veterinary journal. La revue veterinaire canadienne 05/2009; 50(4):401-4. · 0.52 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: This study was designed to evaluate the feasibility and limitations of transcatheter embolization coil occlusion of patent ductus arteriosus in dogs using a carotid artery approach. Seven dogs examined at the University of California, Davis Veterinary Medical Teaching Hospital in 2002-2003 for evaluation of heart disease had congenital patent ductus arteriosus diagnosed by characteristic physical, electrocardiographic, radiographic, and anatomic and Doppler echocardiographic findings. Dogs were anesthetized for transesophageal echocardiography and transcatheter coil embolization of the ductus via the right external carotid artery. Coil embolization was achieved in all seven cases, using one to four detachable embolization coils. There were no major complications. Minor complications occurred in two dogs (additional coils placed using a femoral arterial approach and coil embolization of a left femoral artery branch). One dog was examined only 24 h post-operatively and had no murmur and trivial residual ductal flow by Doppler echocardiography. The other 6 dogs were clinically healthy when examined up to three years post-intervention. One dog had a very soft continuous murmur and mild residual ductal flow; the other five had no audible continuous murmur, with only one dog having trivial residual ductal flow identified by Doppler echocardiography. Although technically challenging, coil embolization via the carotid artery is a viable alternative approach for transcatheter closure of patent ductus arteriosus in some dogs.Journal of veterinary cardiology: the official journal of the European Society of Veterinary Cardiology 10/2009; 11(2):129-36. DOI:10.1016/j.jvc.2009.09.001 · 1.32 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: Closure of reversed patent ductus arteriosus (PDA) is generally accepted to be contraindicated due to case based evidence of worsened outcomes, but little is known about closure of left-to-right PDA with concurrent pulmonary hypertension (PH). This report describes three dogs presenting with varying severity of PH and clinical signs, all with documented left-to-right PDA. The PDA was closed in each case; either by surgical ligation or transarterial device occlusion, and follow up was available for a minimum of 8 months. Every case had a successful outcome with improvement or resolution of PH and associated clinical signs.Journal of veterinary cardiology: the official journal of the European Society of Veterinary Cardiology 02/2010; 12(1):67-73. DOI:10.1016/j.jvc.2009.12.002 · 1.32 Impact Factor