Determinants of gradient field-induced current in a pacemaker lead system in a magnetic resonance imaging environment
ABSTRACT The determinants of low-frequency-induced current by magnetic resonance imaging (MRI) gradient fields in a pacemaker lead system are largely unknown.
The purpose of this study was to determine the magnitude of MRI low-frequency-induced current in an implanted pacemaker lead system and to investigate in vivo determinants of low-frequency-induced current in an animal model.
Six mongrel dogs underwent conventional single-chamber pacemaker implantation with a current recorder connected in series. Pulse generator (PG) was programmed to VOO 120 bpm with subthreshold output. MRI was performed in a 1.5-T scanner. Low-frequency-induced current was recorded during unipolar pacing, bipolar pacing, and bipolar pacing with the PG case electrically isolated from the pocket. In each mode, low-frequency-induced current was recorded with and without a large loop of additional lead connected in series.
With a conventional implant, low-frequency-induced current was < or =0.5 mA in all three pacing modes. With five external loops, the magnitude of low-frequency-induced current increased to >30 mA, with consistent myocardial capture in unipolar and bipolar pacing. However, in bipolar pacing with the PG electrically isolated from the pocket, low-frequency-induced current decreased to <0.5 mA with no myocardial capture even with additional looped leads.
Under conventional implant conditions, the magnitude of low-frequency-induced current is <0.5 mA and is unlikely to cause myocardial capture; however, arrhythmia induction cannot be excluded. With sufficient increase in effective loop area (additional looped leads), direct myocardial capture by the low-frequency-induced current is possible. In this study, breaking the return pathway by electrically isolating the PG case from the circuit abolished low-frequency-induced current.
- SourceAvailable from: Ergin Atalar
[Show abstract] [Hide abstract]
- "However, in  and , no experimental verification is performed and in , experiments are only performed for the Helmholtz coil. On the other hand, in , experiments were performed on six mongrel dogs and the induced current was measured with a current recorder. However, no analytical explanation about the stimulation risk is carried out. "
ABSTRACT: During magnetic resonance imaging, there is an interaction between the time-varying magnetic fields and the active implantable medical devices (AIMD). In this study, in order to express the nature of this interaction, simplified analytical expressions for the electric fields induced by time-varying magnetic fields are derived inside a homogeneous cylindrical volume. With these analytical expressions, the gradient induced potential on the electrodes of the AIMD can be approximately calculated if the position of the lead inside the body is known. By utilizing the fact that gradient coils produce linear magnetic field in a volume of interest, the simplified closed form electric field expressions are defined. Using these simplified expressions, the induced potential on an implant electrode has been computed approximately for various lead positions on a cylindrical phantom and verified by comparing with the measured potentials for these sample conditions. In addition, the validity of the method was tested with isolated frog leg stimulation experiments. As a result, these simplified expressions may help in assessing the gradient-induced stimulation risk to the patients with implants.IEEE transactions on bio-medical engineering 08/2012; 59(10):2845-51. DOI:10.1109/TBME.2012.2212190 · 2.23 Impact Factor
- Heart rhythm: the official journal of the Heart Rhythm Society 12/2008; 5(11):e2; author reply e2-3. DOI:10.1016/j.hrthm.2008.06.013 · 4.92 Impact Factor
- Heart rhythm: the official journal of the Heart Rhythm Society 02/2009; 6(1):138-43. DOI:10.1016/j.hrthm.2008.10.021 · 4.92 Impact Factor