Exposure Assessment Branch, Health Effects Laboratory Division (HELD), National Institute for Occupational Safety and Health (NIOSH), Centers for Disease Control and Prevention (CDC), Morgantown, WV 26505, USA.
A laboratory study was performed to measure quartz in coal dust using high-flow rate samplers (CIP10-R, GK2.69 cyclone, and FSP10 cyclone) and low-flow rate samplers [10-mm nylon and Higgins-Dewell type (BGI4L) cyclones] and to determine whether an increased mass collection from high-flow rate samplers would affect the subsequent quartz measurement by Fourier transform infrared (FTIR) and X-ray diffraction (XRD) analytical procedures. Two different sizes of coal dusts, mass median aerodynamic diameter 4.48 μm (Coal Dust A) and 2.33 μm (Coal Dust B), were aerosolized in a calm air chamber. The mass of coal dust collected by the samplers was measured gravimetrically, while the mass of quartz collected by the samplers was determined by FTIR (NIOSH Manual of Analytical Method 7603) and XRD (NIOSH Manual of Analytical Method 7500) after one of two different indirect preparations. Comparisons between high-flow rate samplers and low-flow rate samplers were made by calculating mass concentration ratios of coal dusts, net mass ratios of coal dusts, and quartz net mass. Mass concentrations of coal dust from the FSP10 cyclone were significantly higher than those from other samplers and mass concentrations of coal dust from 10-mm nylon cyclone were significantly lower than those from other samplers, while the CIP10-R, GK2.69, and BGI4L samplers did not show significant difference in the comparison of mass concentration of coal dusts. The BGI4L cyclone showed larger mass concentration of ∼9% compared to the 10-mm nylon cyclone. All cyclones provided dust mass concentrations that can be used in complying with the International Standard Organization standard for the determination of respirable dust concentration. The amount of coal dust collected from the high-flow rate samplers was found to be higher with a factor of 2-8 compared to the low-flow rate samplers but not in direct proportion of increased flow rates. The high-flow rate samplers collected more quartz compared to low-flow rate samplers in the range of 2-10. There was no significant difference between the per cent (%) quartz in coal dust between the FTIR and XRD analyses. The findings of this study indicated that the increased mass of quartz collected with high-flow rate samplers would provide precise analytical results (i.e. significantly above the limit of detection and/or limit of quantification) compared to the mass collected with low-flow rate samplers, especially in environments with low concentrations of quartz or where short sampling times are desired.
"The difficulties of obtaining accurate measurements for short-term sampling (<4h) or measuring air concentrations of airborne particles <0.05 mg·m−3 with air samplers of flow rates <4 l · min−1 routinely employed for collecting RCS have been discussed (Stacey, 2007). To meet the emerging needs, newer samplers were investigated that can run at higher flow rates >4 l · min−1 (Stacey and Thorpe, 2010; Lee et al., 2010, 2012). If Occupational Hygienists are to use such samplers routinely then it is important to understand the relative differences between emerging and current sampler designs. "
[Show abstract][Hide abstract] ABSTRACT: Prolonged exposure to respirable crystalline silica (RCS) causes silicosis and is also considered a cause of cancer. To meet emerging needs for precise measurements of RCS, from shorter sampling periods (<4h) and lower air concentrations, collaborative work was done to assess the differences between personal respirable samplers at higher flow rates. The performance of FSP10, GK2.69, and CIP 10 R samplers were compared with that of the Safety In Mines Personal Dust Sampler (SIMPEDS) sampler as a reference, which is commonly used in the UK for the measurement of RCS. In addition, the performance of the FSP10 and GK 2.69 samplers were compared; at the nominal flow rates recommended by the manufacturers of 10 and 4.2 l · min(-1) and with flow rates proposed by the National Institute for Occupational Safety and Health of 11.2 and 4.4 l · min(-1). Samplers were exposed to aerosols of ultrafine and medium grades of Arizona road dust (ARD) generated in a calm air chamber. All analyses for RCS in this study were performed at the Health and Safety Laboratory. The difference in flow rates for the GK2.69 is small and does not result in a substantial difference in collection efficiency for the dusts tested, while the performance of the FSP10 at 11.2 l · min(-1) was more comparable with samples from the SIMPEDS. Conversely, the GK2.69 collected proportionately more crystalline silica in the respirable dust than other samplers, which then produced RCS results most comparable with the SIMPEDS. The CIP 10 R collected less ultrafine ARD than other samplers, as might be expected based on earlier performance evaluations. The higher flow rate for the FSP10 should be an added advantage for task-specific sampling or when measuring air concentrations less than current occupational exposure limits.
[Show abstract][Hide abstract] ABSTRACT: Kaolinite interferes with the infrared analysis of quartz. Improper correction can cause over- or underestimation of silica concentration. The standard sampling method for quartz in coal mine dust is size selective, and, since infrared spectrometry is sensitive to particle size, it is intuitively better to use the same size fractions for quantification of quartz and kaolinite. Standard infrared spectrometric methods for quartz measurement in coal mine dust correct interference from the kaolinite, but they do not specify a particle size for the material used for correction. This study compares calibration curves using as-received and respirable size fractions of nine different examples of kaolinite in the different correction methods from the National Institute for Occupational Safety and Health Manual of Analytical Methods (NMAM) 7603 and the Mine Safety and Health Administration (MSHA) P-7. Four kaolinites showed significant differences between calibration curves with as-received and respirable size fractions for NMAM 7603 and seven for MSHA P-7. The quartz mass measured in 48 samples spiked with respirable fraction silica and kaolinite ranged between 0.28 and 23% (NMAM 7603) and 0.18 and 26% (MSHA P-7) of the expected applied mass when the kaolinite interference was corrected with respirable size fraction kaolinite. This is termed "deviation," not bias, because the applied mass is also subject to unknown variance. Generally, the deviations in the spiked samples are larger when corrected with the as-received size fraction of kaolinite than with the respirable size fraction. Results indicate that if a kaolinite correction with reference material of respirable size fraction is applied in current standard methods for quartz measurement in coal mine dust, the quartz result would be somewhat closer to the true exposure, although the actual mass difference would be small. Most kinds of kaolinite can be used for laboratory calibration, but preferably, the size fraction should be the same as the coal dust being collected.
Journal of Occupational and Environmental Hygiene 08/2013; 10(8):425-34. DOI:10.1080/15459624.2013.801819 · 1.17 Impact Factor
[Show abstract][Hide abstract] ABSTRACT: SUMMARY The objective of the present study is to quantify quartz mass in coal dust deposited on internal cassette surface of respirable size-selective samplers. Coal dust was collected with four different respirable size-selective samplers (10 mm Dorr-Oliver nylon, SKC Aluminum, BGI4L, and GK2.69 cyclones) with two different cassette types (polystyrene and static-dissipative polypropylene cassettes). The coal dust was aerosolized in a calm air chamber by using a fluidized bed aerosol generator without neutralization under the assumption that the procedure is similar to field sampling conditions. The mass of coal dust was measured gravimetrically and quartz mass was determined by Fourier transform infrared spectroscopy according to NIOSH Manual of Analytical Method 7603. The mass fractions of the total quartz sample on the internal cassette surface are significantly different between polystyrene and static-dissipative cassettes for all cyclones (p<0.05). No consistent relationship between quartz mass on cassette internal surface and coal dust filter mass was observed. The BGI4L cyclone showed larger (but not significantly) and the GK2.69 cyclone showed significantly lower (p<0.05) internal surface deposit quartz mass fraction for polystyrene cassettes compared to other cyclones. This study confirms previous observations that the interior surface deposits in polystyrene cassettes attached to cyclone pre-selectors can be a substantial part of the sample, and therefore need to be included in any analysis for accurate exposure assessment. On the other hand, the research presented here supports the position that the internal surface deposits in static-dissipative cassettes used with size-selective cyclones are negligible and that it is only necessary to analyze the filter catch.
Journal of Occupational and Environmental Hygiene 09/2014; 11(12). DOI:10.1080/15459624.2014.960575 · 1.17 Impact Factor
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.