Velocity profiles and shear strain rate variability in the USP Dissolution Testing Apparatus 2 at different impeller agitation speeds.
ABSTRACT The fluid velocity profiles at different locations inside a standard USP Dissolution Testing Apparatus 2 were experimentally obtained via Laser Doppler Velocimetry (LDV) at three impeller agitations speeds, namely 50rpm, 75rpm and 100rpm. The experimental results were compared with the predictions obtained with Computational Fluid Dynamics (CFD) where the κ-ω model with low Reynolds number correction was used to account for turbulence effects. In general, good agreement was found between the experimental LDV velocity measurements and the CFD simulation predictions. The non-dimensional tangential, axial and radial velocity profiles (scaled with the impeller tip speed) and the flow pattern were found to be nearly independent of the agitation speed in most regions of the vessel, implying that increasing the agitation speed generally produced a corresponding increase in the local values of the velocity. However, the velocity profiles and flow pattern in the inner core region just below the impeller, where the dissolving tablet is usually located, were found to be much less sensitive to agitation speed. In this region, the axial and radial velocities were especially low and were not significantly affected by agitation increases. This inner core region at the center of the vessel bottom persisted irrespective of agitation intensity. The CFD predictions also indicated that increasing the agitation speed resulted in a higher shear strain rate distribution along the vessel bottom, although the strain rate was always very low at the center of the vessel bottom, even when the agitation speed was increased.
- Journal of Pharmaceutical Sciences 11/1967; 56(10):1238-42. · 3.13 Impact Factor
- [show abstract] [hide abstract]
ABSTRACT: The calibrator tablets now used in the USP suitability test do not reveal common sources of systematic error associated with Apparatus 2. When the apparatus was operated under conditions near or beyond USP tolerances, changes in the results of the USP calibrators were slight, whereas those of several samples of commercial prednisone tablets were significant. Thus, the USP calibrators and requirements do not guarantee suitability of the equipment for general dissolution testing of drug products.Journal of Pharmaceutical Sciences 09/1983; 72(8):910-3. · 3.13 Impact Factor
- Journal of Pharmaceutical Sciences 05/1982; 71(4):451-2. · 3.13 Impact Factor