Enhancement of sensitivity of electrochemical heavy metal detection by evaporative concentration using a super-hydrophobic surface
Graduate Sch. of Pure & Appl. Sci., Tsukuba Univ., JapanDOI: 10.1109/SENSOR.2005.1497295 Conference: Solid-State Sensors, Actuators and Microsystems, 2005. Digest of Technical Papers. TRANSDUCERS '05. The 13th International Conference on, Volume: 2
Source: IEEE Xplore
A novel method to concentrate a droplet of a sample solution on a chip evaporatively and improve the sensitivity for electrochemical heavy metal detection was developed. The chip consisted of a working electrode and a Ag/AgCl reference electrode surrounded by a super-hydrophobic surface. When a droplet of an aqueous solution placed on the area was evaporated, it shrank to the sensitive area making contact with the sensitive area. This effectively concentrated the analyte heavy metal ions. Anodic stripping voltammetry was conducted using the system and the technique. To analyze various heavy metal ions, a bismuth or gold electrode was used as the working electrode. A significant increase in peak height was observed by the evaporative concentration, and the effect was enhanced as the volume of the solution increased and the area of the sensitive area decreased.
- [Show abstract] [Hide abstract]
ABSTRACT: Over millions of years plant surfaces evolved optimized complex multifunctional interfaces. They fulfill different functions in terrestrial plants such as limitation of uncontrolled water loss, protection against various biotic and abiotic influences, and they play a role in the attachment of insects. A recent overview on plant surface functions is presented by Jeffree (in Riederer, 2006). One of the most remarkable functions is closely linked with plant epicuticular waxes. The outermost barrier is formed by a cuticle consisting of two major components: a polyester matrix with embedded and overlaying lipids.Functional Surfaces in Biology, 12/2008: pages 97-111;
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.