Article

# Uniform precision ultrasound strain imaging.

Departmentof Engineering, University of Cambridge, Cambridge, UK.

IEEE transactions on ultrasonics, ferroelectrics, and frequency control (Impact Factor: 1.8). 11/2009; 56(11):2420-36. DOI: 10.1109/TUFFc.2009.1330 Source: PubMed

- [Show abstract] [Hide abstract]

**ABSTRACT:**This paper introduces two real-time elastography techniques based on analytic minimization (AM) of regularized cost functions. The first method (1D AM) produces axial strain and integer lateral displacement, while the second method (2D AM) produces both axial and lateral strains. The cost functions incorporate similarity of radio-frequency (RF) data intensity and displacement continuity, making both AM methods robust to small decorrelations present throughout the image. We also exploit techniques from robust statistics to make the methods resistant to large local decorrelations. We further introduce Kalman filtering for calculating the strain field from the displacement field given by the AM methods. Simulation and phantom experiments show that both methods generate strain images with high SNR, CNR and resolution. Both methods work for strains as high as 10% and run in real-time. We also present in vivo patient trials of ablation monitoring. An implementation of the 2D AM method as well as phantom and clinical RF-data can be downloaded.IEEE transactions on medical imaging. 11/2010; 30(4):928-45. - [Show abstract] [Hide abstract]

**ABSTRACT:**Ultrasound elastography is a technique used for clinical imaging of tissue stiffness with a conventional ultrasound machine. It was first proposed two decades ago, but active research continues in this area to the present day. Numerous clinical applications have been investigated, mostly related to cancer imaging, and though these have yet to prove conclusive, the technique has seen increasing commercial and clinical interest. This paper presents a review of the most widely adopted, non-quantitative, techniques focusing on technical innovations rather than clinical applications. The review is not intended to be exhaustive, concentrating instead on placing the various techniques in context according to the authors' perspective of the field.Interface focus: a theme supplement of Journal of the Royal Society interface 08/2011; 1(4):540-52. · 2.21 Impact Factor - [Show abstract] [Hide abstract]

**ABSTRACT:**Axial displacement estimation is fundamental to many freehand quasistatic ultrasonic strain imaging systems. In this paper, we present a novel estimation method that combines the strengths of quality-guided tracking, multi-level correlation, and phase-zero search to achieve high levels of accuracy and robustness. The paper includes a full description of the hybrid method, in vivo examples to illustrate the methodÂ¿s clinical relevance, and finite element simulations to assess its accuracy. Quantitative and qualitative comparisons are made with leading single- and multi-level alternatives. In the in vivo examples, the hybrid method produces fewer obvious peak-hopping errors, and in simulation, the hybrid method is found to reduce displacement estimation errors by 5 to 50%. With typical clinical data, the hybrid method can generate more than 25 strain images per second on commercial hardware; this is comparable with the alternative approaches considered in this paper.IEEE Transactions on Ultrasonics Ferroelectrics and Frequency Control 05/2010; · 1.82 Impact Factor

Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.