Noise and Signal Analysis of Ir/Au TES With Asymmetrical Slits Parallel to the Electric Current
Nat. Inst. of Adv. Ind. Sci. & Technol., Ibaraki, JapanIEEE Transactions on Applied Superconductivity (Impact Factor: 1.24). 07/2005; 15(2):522 - 525. DOI: 10.1109/TASC.2005.849896
Source: IEEE Xplore
A multi-pixel TES array is one of the attractive methods to read-out the incident x-ray position, which simplifies a read-out circuit for imaging applications. We fabricated a ten-pixel Ir/Au TES microcalorimeter with asymmetrical slits parallel to the electric current, and analyzed its noise and signal characteristics. The device was successfully operated in a strong electro-thermal feedback (ETF) mode. However, very large excess noise is observed, which cannot be explained by the noise theory for a single pixel TES. We have modeled the multi-pixel TES array as separated thermal group components, and analyzed noise performance theoretically. As a result, the current noise in our theory is well agreed with the measured noise at a low frequency below 3 kHz. From the x-ray irradiation measurements, ten divided signal groups are observed, which will be associated with the incident x-ray position pixel. The best energy resolution was 18 eV (FWHM) for 5.9 keV x-rays at 0.5 μV bias voltage.
Article: Characterization of Ir/Au pixel TES[Show abstract] [Hide abstract]
ABSTRACT: Signal shapes and noise characteristics of an asymmetrical ten-pixel Ir/Au-TES have been studied. The asymmetric design may be effective to realize an imaging spectrometer. Distinct two exponential decays observed for X-ray events are consistent with a two-step R–T curve. A theoretical thermal model for noise in multi-pixel devices reasonably explains the experimental data.Nuclear Instruments and Methods in Physics Research Section A Accelerators Spectrometers Detectors and Associated Equipment 04/2006; 559(2):429-431. DOI:10.1016/j.nima.2005.12.029 · 1.22 Impact Factor
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.