Proton Therapy for Prostate Cancer Treatment Employing Online Image Guidance and an Action Level Threshold
ABSTRACT The ability to determine the accuracy of the final prostate position within a determined action level threshold for image-guided proton therapy is unclear.
Three thousand one hundred ten images for 20 consecutive patients treated in 1 of our 3 proton prostate protocols from February to May of 2007 were analyzed. Daily kV images and patient repositioning were performed employing an action-level threshold (ALT) of > or = 2.5 mm for each beam. Isocentric orthogonal x-rays were obtained, and prostate position was defined via 3 gold markers for each patient in the 3 axes.
To achieve and confirm our action level threshold, an average of 2 x-rays sets (median 2; range, 0-4) was taken daily for each patient. Based on our ALT, we made no corrections in 8.7% (range, 0%-54%), 1 correction in 82% (41%-98%), and 2 to 3 corrections in 9% (0-27%). No patient needed 4 or more corrections. All patients were treated with a confirmed error of < 2.5 mm for every beam delivered. After all corrections, the mean and standard deviations were: anterior-posterior (z): 0.003 +/- 0.094 cm; superior-inferior (y): 0.028 +/- 0.073 cm; and right-left (x) -0.013 +/- 0.08 cm.
It is feasible to limit all final prostate positions to less than 2.5 mm employing an action level image-guided radiation therapy (IGRT) process. The residual errors after corrections were very small.
- SourceAvailable from: Daniel R Simpson[Show abstract] [Hide abstract]
ABSTRACT: Radiotherapy has a well-established role in the management of head and neck cancers. Over the past decade, a variety of new imaging modalities have been incorporated into the radiotherapy planning and delivery process. These technologies are collectively referred to as image-guided radiotherapy and may lead to significant gains in tumor control and radiation side effect profiles. In the following review, these techniques as they are applied to head and neck cancer patients are described, and clinical studies analyzing their use in target delineation, patient positioning, and adaptive radiotherapy are highlighted. Finally, we conclude with a brief discussion of potential areas of further radiotherapy advancement.Journal of Oncology 02/2009; 2009(1687-8450):752135. DOI:10.1155/2009/752135
- [Show abstract] [Hide abstract]
ABSTRACT: To develop a device that allows supine craniospinal proton and photon therapy to the vast majority of proton and photon facilities currently experiencing limitations as a result of couch design issues. Plywood and carbon fiber were used for the development of a prototype unit. Once this was found to be satisfactory after all design issues were addressed, computer-assisted design (CAD) was used and carbon fiber tables were built to our specifications at a local manufacturer of military and racing car carbon fiber parts. Clinic-driven design was done using real-time team discussion for a prototype design. A local machinist was able to construct a prototype unit for us in <2 weeks after the start of our project. Once the prototype had been used successfully for several months and all development issues were addressed, a custom carbon fiber design was developed in coordination with a carbon fiber manufacturer in partnership. CAD methods were used to design the units to allow oblique fields from head to thigh on patients up to 200 cm in height. Two custom-designed carbon fiber craniospinal tabletop designs now exist: one long and one short. Four are in successful use in our facility. Their weight tolerance is greater than that of our robot table joint (164 kg). The long unit allows for working with taller patients and can be converted into a short unit as needed. An affordable, practical means of doing supine craniospinal therapy with protons or photons can be used in most locations via the use of these devices. This is important because proton therapy provides a much lower integral dose than all other therapy methods for these patients and the supine position is easier for patients to tolerate and for anesthesia delivery. These units have been successfully used for adult and pediatric supine craniospinal therapy, proton therapy using oblique beams to the low pelvis, treatment of various spine tumors, and breast-sparing Hodgkin's therapy.Medical dosimetry: official journal of the American Association of Medical Dosimetrists 08/2012; 38(1). DOI:10.1016/j.meddos.2012.07.003 · 0.95 Impact Factor