A Two-Pulse Technique for Extracting 3(rd) Harmonic from Ulasound Contrast Agent Echo Signal
Conference proceedings: ... Annual International Conference of the IEEE Engineering in Medicine and Biology Society. IEEE Engineering in Medicine and Biology Society. Conference 02/2008; 2008:5415-8. DOI: 10.1109/IEMBS.2008.4650439
Multi-pulse techniques like CPS (contrast pulse sequence) and TPS (triplet pulse sequence) are the most popular methods for separating the 3rd harmonic signals from received signal. Those two methods, however, transmit a pulse at least three times along each scanline with different phase and amplitude, which results in the frame rate reduction. In this paper, we propose a technique using two pulses whose phase difference is 90 degrees and a simple digital filter. The second harmonic signal is eliminated by summing two received signals as their phase difference becomes 180 degrees and then the fundamental signals are eliminated by using a digital filter. Computer simulations are performed for different values of signal bandwidths and filter specifications. The results show the maximum error is -35.5 dB compared to TPS.
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.