Quantitative assessment of levodopa-induced dyskinesia using automated motion sensing technology
Conference proceedings: ... Annual International Conference of the IEEE Engineering in Medicine and Biology Society. IEEE Engineering in Medicine and Biology Society. Conference 08/2012; 2012:154-7. DOI: 10.1109/EMBC.2012.6345894
The objective was to capture levodopa-induced dyskinesia (LID) in patients with Parkinson's disease (PD) using body-worn motion sensors. Dopaminergic treatment in PD can induce abnormal involuntary movements, including choreatic dyskinesia (brief, rapid, irregular movements). Adjustments in medication to reduce LID often sacrifice control of motor symptoms, and balancing this tradeoff poses a significant challenge for management of advanced PD. Fifteen PD subjects with known LID were recruited and instructed to perform two stationary motor tasks while wearing a compact wireless motion sensor unit positioned on each hand over the course of a levodopa dose cycle. Videos of subjects performing the motor tasks were later scored by expert clinicians to assess global dyskinesia using the modified Abnormal Involuntary Rating Scale (m-AIMS). Kinematic features were extracted from motion data in different frequency bands (1-3Hz and 3-8Hz) to quantify LID severity and to distinguish between LID and PD tremor. Receiver operator characteristic analysis was used to determine thresholds for individual features to detect the presence of LID. A sensitivity of 0.73 and specificity of 1.00 were achieved. A neural network was also trained to output dyskinesia severity on a 0 to 4 scale, similar to the m-AIMS. The model generalized well to new data (coefficient of determination= 0.85 and mean squared error= 0.3). This study demonstrated that hand-worn motion sensors can be used to assess global dyskinesia severity independent of PD tremor over the levodopa dose cycle.
Article: Mobile Motion Capture - MiMiC[Show abstract] [Hide abstract]
ABSTRACT: The low cost, simple, robust, mobile, and easy to use Mobile Motion Capture (MiMiC) system is presented and the constraints which guided the design of MiMiC are discussed. The MiMiC Android application allows motion data to be captured from kinematic modules such as Shimmer 2r sensors over Bluetooth. MiMiC is cost effective and can be used for an entire day in a person's daily routine without being intrusive. MiMiC is a flexible motion capture system which can be used for many applications including fall detection, detection of fatigue in industry workers, and analysis of individuals' work patterns in various environments.Conference proceedings: ... Annual International Conference of the IEEE Engineering in Medicine and Biology Society. IEEE Engineering in Medicine and Biology Society. Conference 07/2013; 2013:3435-3438. DOI:10.1109/EMBC.2013.6610280
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.