Conference Paper

Advancing towards smart endoscopy with specific electronics to enable locomotion and focusing capabilities in a wireless endoscopic capsule robot

• A. Dieguez
Dept. of Electron., Univ. of Barcelona, Barcelona, Spain
DOI: 10.1109/BIOCAS.2009.5372047 Conference: Biomedical Circuits and Systems Conference, 2009. BioCAS 2009. IEEE
Source: IEEE Xplore

ABSTRACT The paper reports the electronics used in a new developed wireless endoscopic capsule provided with novel focus and locomotion features. The locomotion is enabled by using legs driven by 2 brushless DC (BLDC) micromotors. The focusing system is enabled by using a liquid lens with a variable focal. These functions are managed by an ASIC that has been developed to provide the specific innovative functions. The size of the ASIC is 3.0 mm Ã 2.7 mm in a 0.35 um high voltage CMOS technology. The ASIC is described in detail, as well as the performances when it is used with the final devices used in the capsule.

0 Bookmarks
·
47 Views
• Source
Article: Lossless and Low-Power Image Compressor for Wireless Capsule Endoscopy.
[Hide abstract]
ABSTRACT: We present a lossless and low-complexity image compression algorithm for endoscopic images. The algorithm consists of a static prediction scheme and a combination of golomb-rice and unary encoding. It does not require any buffer memory and is suitable to work with any commercial low-power image sensors that output image pixels in raster-scan fashion. The proposed lossless algorithm has compression ratio of approximately 73% for endoscopic images. Compared to the existing lossless compression standard such as JPEG-LS, the proposed scheme has better compression ratio, lower computational complexity, and lesser memory requirement. The algorithm is implemented in a 0.18 μm CMOS technology and consumes 0.16 mm × 0.16 mm silicon area and 18 μW of power when working at 2 frames per second.
VLSI Design 01/2011; 2011.
• Article: Subsample-based image compression for capsule endoscopy
[Hide abstract]
ABSTRACT: The key design challenge in capsule endoscopic system is to reduce the area and power consumption while maintaining acceptable video quality. In this paper, a subsample-based image compressor for such endoscopic application is presented. The algorithm is developed around some special features of endoscopic images. It consists of a differential pulse coded modulation followed by Golomb-rice coding. Based on the nature of endoscopic images, several subsampling schemes on the chrominance components are applied. This scheme is particularly suitable to work with any commercial low-power image sensors that outputs image pixels in a raster scan fashion, eliminating the need of memory buffer, as well as temporary storage (as needed in transform coding schemes). An image corner clipping algorithm is also presented. The reconstructed images have been verified by medical doctors for acceptability. The proposed algorithm has a very low complexity and is suitable for the VLSI implementation. Compared to other transform-based algorithms targeted to capsule endoscopy, the proposed raster-scan-based scheme performs very strongly with a compression ratio of 80% and a very high reconstruction PSNR (over 45dB). KeywordsDifferential pulse coded modulation–Capsule endoscopic application–Golomb coding–Image compression
Journal of Real-Time Image Processing 01/2011; · 1.16 Impact Factor
• Article: Low Power and Low Complexity Compressor for Video Capsule Endoscopy.
[Hide abstract]
ABSTRACT: The main challenge in video capsule endoscopic system is to reduce the area and power consumption while maintaining acceptable video reconstruction. In this paper, a subsample-based data compressor for video endoscopy application is presented. The algorithm is developed around the special features of endoscopic images that consists of a differential pulse-coded modulation (DPCM) followed by Golomb–Rice coding. Based on the nature of endoscopic images, several subsampling schemes on the chrominance components are applied. This video compressor is designed in a way to work with any commercial low-power image sensors that outputs image pixels in a raster scan fashion, eliminating the need of memory buffer and temporary storage (as needed in transform coding schemes). An image corner clipping algorithm is also introduced. The reconstructed images have been verified by five medical doctors for acceptability. The proposed low-complexity design is implemented in a 0.18 $\mu{\rm m}$ CMOS technology and consumes 592 standard cells, $0.16~{\rm mm}\times 0.16~{\rm mm}$ silicon area, and 42 $\mu{\rm W}$ of power. Compared to other algorithms targeted to video capsule endoscopy, the proposed raster-scan-based scheme performs strongly with a compression ratio of 80% and a very high reconstruction peak signal-to-noise ratio (over 48 dB).
IEEE Transactions on Circuits and Systems for Video Technology 01/2011; 21:1534-1546. · 1.82 Impact Factor