Efficacy of the Disposcope endoscope, a new video laryngoscope, for endotracheal intubation in patients with cervical spine immobilisation by semirigid neck collar: Comparison with the Macintosh laryngoscope using a simulation study on a manikin
Department of Emergency Medicine, Konkuk University Medical Center, Konkuk University School of Medicine, Seoul, Korea.Emergency Medicine Journal (Impact Factor: 1.84). 04/2012; 30(4). DOI: 10.1136/emermed-2011-200899
OBJECTIVE: To evaluate whether endotracheal intubation in patients with cervical spine immobilisation by a semirigid neck collar is easier using the Disposcope endoscope (DE), a new video laryngoscope, than with the Macintosh laryngoscope (ML). METHODS: Sixty-eight medical interns who participated in a training programme for endotracheal intubation using the DE and ML were recruited to the randomised crossover trial 1 week after completing the training programme. In the trial, they used both the DE and the ML to perform intubation on a manikin wearing a semirigid neck collar. The time required to view the vocal cords and to complete intubation, successful endotracheal intubation, modified Cormack-Lehane classification (CL grade) and dental injury were recorded and analysed. RESULTS: The mean (SD) time to view the vocal cords was significantly shorter with the DE than with the ML (10.0 (7.0) vs 20.8 (18.9) s; p<0.0001). There were higher rates of CL grades 1 and 2a (69.1% and 22.1%) using the DE than with the ML (10.3% and 14.7%). All 68 participants had a higher rate of successful endotracheal intubation using the DE than using the ML (68 (100%) vs 47 (69.1%); p<0.0001). It took less time to complete endotracheal intubation with the DE than with the ML (p<0.0001). CONCLUSIONS: In patients with cervical spine immobilisation by a semirigid neck collar, the DE may be a more effective device for endotracheal intubation than the ML.
- [Show abstract] [Hide abstract]
ABSTRACT: Endotracheal intubation is a common procedure in the emergency department, and new devices may improve intubation time, success, or view. We compared the King Vision video laryngoscope (KVVL; King Systems, Noblesville, IN) to the Macintosh direct laryngoscope (DL) in simulated normal and difficult airways. Using manikins and clinical-grade cadavers, difficult airway scenarios were simulated using head movement restriction or a cervical spine collar. Four scenarios were studied using the KVVL and DL: normal manikin airway, difficult manikin airway, normal cadaver airway, and difficult cadaver airway. Primary outcomes were time to intubation and rate of successful intubation. Secondary outcomes were the percent of glottic opening and Cormack-Lehane grade visualized. Thirty-two paramedics participated in the study. In the normal manikin airway scenario, time to intubation was 3.4 s (99% confidence interval [CI] 0.1-6.6) faster with the KVVL compared with DL. Time to intubation was 11.3 s (99% CI 2.4-20.2) faster with the KVVL in the difficult cadaver airway scenario. There was no difference in time to intubation in the other 2 scenarios. In the difficult cadaver airway, 10 of 32 participants failed to successfully intubate the trachea using DL, whereas all KVVL intubations were successful. All scenarios found a lower Cormack-Lehane grade and higher percentage of glottic opening with the KVVL compared to DL. The KVVL was slightly faster than Macintosh DL in two of four studied airway scenarios, and had a higher success rate in the difficult cadaver airway scenario. Further study is required in the clinical setting.Journal of Emergency Medicine 04/2014; 47(2). DOI:10.1016/j.jemermed.2014.02.008 · 0.97 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: We aimed to study the feasibility of a comfortable and secure intubation achieved with the Disposcope endoscope or Macintosh laryngoscope when glottis viewing is difficult. Forty adults scheduled for elective surgery under general anesthesia, in whom glottis viewing was difficult with the Macintosh laryngoscope (classified as Cormack-Lehane Grade III or IV), were randomized into 2 groups (N = 20 each): Disposcope endoscope (Group D) and Macintosh laryngoscope (Group M). We recorded the successful glottis viewing rate; intubation time; successful intubation rate; incidence of systolic blood pressure (SBP) ≥140 mmHg and heart rate (HR) ≥90 bpm from the beginning of intubation to 5 min after intubation; and postoperative sore throat and hoarseness. Successful glottis viewing rate and successful first intubation rate were higher in Group D than in Group M (P < 0.05); the number of intubations taking >3 min and with SBP ≥140 mmHg and HR ≥90 bpm were less in Group D (P < 0.05). The visual analogue scale of postoperative sore throat 1 day after extubation was higher in Group M than in Group D (P < 0.05). No significant differences were found in other indices (P > 0.05). Better stability of hemodynamics, less intubation time, higher successful first intubation rate, and fewer incidences of postoperative sore throat were achieved in Group D than in Group M. Thus, comfortable and secure intubation can be achieved using the Disposcope endoscope in patients in whom glottis viewing is difficult.Genetics and molecular research: GMR 04/2015; 14(2):3694-3701. DOI:10.4238/2015.April.17.19 · 0.78 Impact Factor
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.