Amperometric biosensor based on carbon nanotubes coated with polyaniline/dendrimer-encapsulated Pt nanoparticles for glucose detection
ABSTRACT A novel amperometric glucose biosensor based on the nanocomposites of multi-wall carbon nanotubes (CNT) coated with polyaniline (PANI) and dendrimer-encapsulated Pt nanoparticles (Pt-DENs) is prepared. CNT coated with protonated PANI is in situ synthesized and Pt-DENs is absorbed on PANI/CNT composite surface by self-assembly method. Then Glucose oxidase (GOx) is crosslink-immobilizated onto Pt-DENs/PANI/CNT composite film. The results show that the fabricated GOx/Pt-DENs/PANI/CNT electrode exhibits excellent response performance to glucose, such as low detection limit (0.5 µM), wide linear range (1 µM–12 mM), short response time (about 5 s), high sensitivity (42.0 µA mM− 1 cm− 2) and stability (83% remains after 3 weeks).
- SourceAvailable from: Bilal Demir[Show abstract] [Hide abstract]
ABSTRACT: This is a PDF file of an unedited manuscript that has been accepted for publication. As a service to our customers we are providing this early version of the manuscript. The manuscript will undergo copyediting, typesetting, and review of the resulting proof before it is published in its final form. Please note that during the production process errors may be discovered which could affect the content, and all legal disclaimers that apply to the journal pertain.Sensors and Actuators B Chemical 05/2013; · 3.84 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: In this paper a highly sensitive glucose biosensor is proposed based on a polysilicon (poly-Si) wire structure coated with 3-aminopropyltriethoxysilane (γ-APTES) mixed with polydimethylsiloxane-treated hydrophobic fumed silica nanoparticles (NPs) as the sensing membrane. The γ-APTES and fumed silica NPs mixture was directly transferred to and coated onto the poly-Si wire region with the help of a focus-ion-beam (FIB) processed capillary atomic-force-microscope (C-AFM) tip. After the necessary curing and UV illumination processes, the resultant sensor showed an extremely wide linear detection range from 0.1μM to 10mM with a channel current sensitivity as high as 5.33AmM−1cm−2 (or a channel conductance sensitivity of 70μSmM−1), and a detection limit as low as 10nM can be achieved. Our experimental results showed that the poly-Si wire sensor has good selective analysis and operational stability on glucose detection under a 10:1 concentration ratio of glucose and uric acid. Its linear range and lowest detection limit remain virtually unimpaired in the presence of uric acid.Sensors and Actuators B-chemical - SENSOR ACTUATOR B-CHEM. 01/2009; 142(1):273-279.
- [Show abstract] [Hide abstract]
ABSTRACT: Nitrogen-doped carbon nanotubes (N-CNTs)/polyaniline (PANI) composites are developed as an electrode material for biosensors. The morphology, composition, and optical properties of the resulting products were characterized by transmission electron microscopy (TEM), thermogravimetric analysis (TGA), Fourier transform infrared spectroscopy (FT-IR), and ultraviolet-visible absorption spectra (UV-vis). Furthermore, N-CNTs/PANI composite was immobilized on the surface of a glassy carbon electrode (GCE) and applied to construct a sensor. The obtained N-CNTs/PANI-modified GCE showed one pair of redox peaks and high catalytic activity for the oxidation of dopamine (DA) in a neutral environment. Differential pulse voltammograms results illustrate that the fabricated DA biosensor has high anti-interference ability towards ascorbic acid (AA). In addition, the fabricated biosensor showed superior performances with two wide linear ranges from 1 to 80 μM and from 1.5 to 3.5 mM and a low detection limit of 0.01 μM.Science China-Chemistry · 1.52 Impact Factor